Marine 15PPM Bilge Alarm Oil Water Monitor
Marine 15PPM Bilge Alarm device is designed in accordance with the
International Maritime Organization MEPC. (49) resolution 107,
approved by the China classification society (CCS) , is the latest
generation monitiring instrument, applicable to ships, continuous
monitoring of ship barge oil, with functions of 15 PPM transfinite
alarm and control of oil conent. It is usually used with 15 PPM
bilge oily water separator.
Marine 15PPM Bilge Alarm is applied the technology of near infrared
measurement,so as to display the detected result in real time. The
unit is supplied with 2 works-adjusted alarms. The set point (15ppm
factory set) could be adjusted at any moment through the panel. If
an alarm set point is exceed, the alarms are visible at the front
panel and the appropriate relays are switched, that can be achieved
the movement control of overrun closedown or the treatment of
overrun recycle, and the memory and enquiry of alarm record.
Furthermore, a 4 - 20 mA (equal to 0 – 15ppm) effective signal
output is available for driving a recorder or external meter.
1) Run oil free water through the instrument to purge the system.
2) Adjust the flow rate through the unit by using the small screws
in the cell cap. Taking out a screw will increase the flow rate.
NB: The flow rate should be checked on both, the clean water supply
and the separator sample supply. If the clean water supply is
obtained from a high pressure source, the flow rate will be higher
than from the sample point. The flow rate is not influencing the
accuracy of the instrument. The adjustment is only important for
the time delay between the sample point and the monitor.
3) Switch on the instrument and make sure, that the Power LCD is
illuminated and it displays the number “EE”, with the fault LED on.
Allow a few minutes for the system to warm up and stabilize. It
works when the fault LED off and measure value displays.
4) The monitor will display "0" with clean water; if the value is
higher than 0, it means there is air mixed up. Then it is necessary
to check the entrance components and sample water pipeline and
reset the position.
5) Through the front control panel offset to operate the display
Marine 15PPM Bilge Alarm Main Performance Indicator
|Range||0-30mg/l (up to 50mg/l)|
|Voltage||220VAC/110VAC ±10% 50Hz/60Hz 5%|
|Alarm output||Two-way relay output|
|Alarm Display||Yellow LED|
Please feel free to contact with us at: sales at weitongmarine dot
com for more details.
Zhuhai Weitong Marine Equipment Co., Ltd
Contact: sales【at】weitongmarine 【dot】com